YSBN-200C JCPT Medical center Infant Care Baby Toddler Radiant Warmer
Microcontroller control open up kind radiant warmer with manual and servo technique
Display pores and skin established facility, skin observed temperature in ºC and warmth power independently
User friendly delicate contact panel massive and vibrant LED Screen
Ceramic infrared heating aspect for successful warming
Audible and visual alarm facility for overheating over and above established temperature range,
probe failure, power failure, program failure and heater failure
Warmer will be feeling the skin probe failure and reduce off the heater
Warmer head will rotate and swivel in diverse direction
Inbuilt battery back again- up in the course of power failure
Keyboard locking facility to avoid undesired consumer modification of the established parameter.
Height of the warmer will adjustable
Control panel will liquid proof and allow easy and hygienic disinfection
JCPT alarms for patient basic safety
Large mattress, drop down aspect panels for simple accessibility to new-born
Examination lamps for easy observation of new-born
Observation JCPT : 1000 Lux
Color Temperature assortment : 3700k to 5100k
Patient management ( baby manner ) : 34°C to 37.5°C in increments .1°C ( 95°F to ninety nine.5°F )
Manual mode alter heater output : 10 – a hundred% with 5% increment, auditory and visual alarm will give each and every 15 min.
Temperature measurement accuracy: ± .3°C, resolution: ± .1°C, probe accuracy: ± .1°C
Warmer have with bassinet trolley. Mattress foam density 25kg/cm³, transparent collapsible side wall, easy detachable for cleaning, mattress size : 20″ x 28 “
Size of fall down 5″ (±1″ ) mattress surface area, 6mm thick, obvious and transparent.
Bed peak about eighty ~ 100cm from the flooring and eighty ~ 90cm from the warmth source.
The infant bed is crevice totally free for simplicity of cleaning, infection management and mattress employed to be
Heater Component : 600 watts
Heater will minimize off , if the optimum irradiance at any stage of the mattress location exceeds a total irradiance amount of 10mw/cm² or over 60% of the heater output ( amongst 10 to thirty min)
Bed tilt mechanism : -10-15°
Skin temperature probe measurement : ≤ 10mm dia three~ 4mm thick ( biocompatible), pliable probe wire and non-stiff
Castor wheels 2 are lockable
Ambient temperature : ten-40°C
Relative Humidity : 15 – 90%
Power supply : 220-240VAC, 50Hz
2.IV pole-one No.
3.Skin Probe-1 No.
four.X-ray cassette tray-one No
5.Operational and support manual – one set every single